Save interesting pages while you browse the DKSH website via the basket icon on the bottom of the page. You will find the pages you saved in this area again.
Save interesting documents while you browse the DKSH website via the basket icon next to the download links. You will find the documents you saved in this area again.
Please note: This information is saved in a cookie. In case your browser deletes cookies after a session, the information will be lost.
Do you really want to delete all documents?
This action can not be undone.
Please leave a message
Eileen Choo, Assistant General Manager
Salt content is determined by measuring the conductivity of a solution of crude oil in a polar solvent when subjected to an alternating electrical current and is obtained by comparison of the resulting conductance to a calibration curve of known salt mixtures.
Determines the salt content, conductance, and temperature of crude oil samples according to ASTM D3230 and IP 265 specifications. Utilizes the latest low-voltage, synchronous detection technology for conductivity measurements and a high-accuracy thermistor array to measure sample temperature. Automatically calculates salt concentration directly from acquired temperature and conductivity values. Measures conductivity over four ranges 0-2, 2-20, 20-200, and 200-1500 mS with automatic range selection. Self-calibration feature allows operator to adjust for any drift without re-entering standard temperature curves. Complete data storage of test results which is limited only by the hard drive capacity of external PC. Easy-to-read alpha-numeric display shows any four of the following parameters at one time as chosen by the operator: salts, conductance, conductance @ 25°C, pH, pH millivolts, temperature (°C or °F), internal and external battery voltages, date, time, logging ID, and ID increment value.